The Multispectral Scanner (MSS) was launched on the Earth Resources Technology Satellite (ERTS-1) 23 July 1972. The MSS has two calibration systems, one internal and one external. Both calibration systems have shown strong, spectrally dependent performance degradation since launch. This paper presents details on the optical system of the MSS and data on the performance degradation as a function of both spectral interval and time in orbit. The history of the MSS during tests is traced, and it is shown that hydrocarbons from an external source may have been deposited on optical surfaces in the instrument. It is postulated that these contaminant coatings may have polymerized as a result of the exposure to uv light from the sun, increasing their blue absorbtion and accounting for the observed performance degradation. Arguments supporting this postulate are presented, and other possible sources of the performance degradation are discussed.
© 1974 Optical Society of America
Original Manuscript: March 26, 1973
Published: May 1, 1974
John J. Horan, Daniel S. Schwartz, and James D. Love, "Partial Performance Degradation of a Remote Sensor in a Space Environment, and Some Probable Causes," Appl. Opt. 13, 1230-1237 (1974)
|Alert me when this paper is cited|
OSA is able to provide readers links to articles that cite this paper by participating in CrossRef's Cited-By Linking service. CrossRef includes content from more than 3000 publishers and societies. In addition to listing OSA journal articles that cite this paper, citing articles from other participating publishers will also be listed.
|Fig. 1||Fig. 2||Fig. 3|
|Fig. 4||Fig. 5||Fig. 6|
|Fig. 7||Fig. 8||Fig. 9|
|Fig. 10||Fig. 11||Fig. 12|