Recently a variety of array detectors have been used in the focal plane of Czerny-Turner plane-grating spectrometers. The output signals have been coupled to a variety of storage units, resulting in optical multichannel analyzers (OMA). We present a clarification of the method for calculating the dispersion of this instrument, as well as a table of useful spectroscopic parameters for spectrometer-detector array combinations.
Leon J. Radziemski, "Calculation of dispersion for a plane grating in a Czerny-Turner mount: a comment," Appl. Opt. 20, 1948-1950 (1981)