## Phase-Shifting Algorithms for Electronic Speckle Pattern Interferometry

Applied Optics, Vol. 41, Issue 1, pp. 46-54 (2002)

http://dx.doi.org/10.1364/AO.41.000046

Acrobat PDF (1978 KB)

### Abstract

A set of innovative phase-shifting algorithms developed to facilitate metrology based on electronic speckle pattern interferometry (ESPI) are presented. The theory of a phase-shifting algorithm, called a (5, 1) algorithm, that takes five phase-shifted intensity maps before a specimen is deformed and one intensity map after a specimen is deformed is presented first. Because a high-speed camera can be used to record the dynamic image of the specimen, this newly developed algorithm has the potential to retain the phase-shifting capability for ESPI in dynamic measurements. Also shown is an algorithm called a (1, 5) algorithm that takes five phase-shifted intensity maps after the specimen is deformed. In addition, a direct-correlation algorithm was integrated with these newly developed (5, 1) or (1, 5) algorithms to form DC-(5, 1) and DC-(1, 5) algorithms, which are shown to improve significantly the quality of the phase maps. The theoretical and experimental aspects of these two newly developed techniques, which can extend ESPI to areas such as high-speed dynamic measurements, are examined in detail.

© 2002 Optical Society of America

**OCIS Codes**

(110.6150) Imaging systems : Speckle imaging

(120.5050) Instrumentation, measurement, and metrology : Phase measurement

(120.6160) Instrumentation, measurement, and metrology : Speckle interferometry

(120.6650) Instrumentation, measurement, and metrology : Surface measurements, figure

**Citation**

Chih-Cheng Kao, Gym-Bin Yeh, Shu-Sheng Lee, Chih-Kung Lee, Ching-Sang Yang, and Kuang-Chong Wu, "Phase-Shifting Algorithms for Electronic Speckle Pattern Interferometry," Appl. Opt. **41**, 46-54 (2002)

http://www.opticsinfobase.org/ao/abstract.cfm?URI=ao-41-1-46

Sort: Year | Journal | Reset

### References

- R. Jones and C. Wykes, Holographic and Speckle Interferometry (Cambridge University, Cambridge, UK, 1983).
- A. J. Moore and J. R. Tyrer, “Phase-stepped ESPI and moire interferometry for measuring the stress-intensity factor and J integral,” Exp. Mech. 35, 306–314 (1995).
- J. N. Butters and J. A. Leendertz, “A double exposure technique for speckle pattern interferometry,” J. Phys. E 4, 277–279 (1971).
- K. Creath, “Phase-shifting speckle interferometry,” Appl. Opt. 24, 3053–3058 (1985).
- P. Hariharan, B. F. Orbel, and T. Eiju, “Digital phase-shifting interferometry: a simple error-compensating phase calculation,” Appl. Opt. 26, 2504–2507 (1987).
- D. R. Schmitt and R. W. Hunt, “Optimization of fringe pattern calculation with direct correlations in speckle interferometry,” Appl. Opt. 36, 8848–8857 (1997).
- C. Gorecki, “Phase-correlation techniques for quasi-real-time measurement of deformations with digital speckle interferometry,” Appl. Opt. 33, 2933–2938 (1994).
- J. Pomarico, R. Arizaga, R. Towoba, and H. Rabal, “Algorithm to compute spacing of digital speckle correlation fringes,” Optik (Stuttgart) 95, 125–127 (1993).
- D. C. Ghiglia and L. A. Romero, “Robust two-dimensional weighted and unweighted phase unwrapping that uses fast transforms and iterative methods,” J. Opt. Soc. Am. A 11, 107–117 (1994).
- H. Takajo and T. Takahashi, “Noniterative method for obtaining the exact solution for the normal equation in a least-squares phase estimation,” J. Opt. Soc. Am. A 5, 1818–1827 (1988).
- H. Takajo and T. Takahashi, “Least-squares phase estimation from the phase difference,” J. Opt. Soc. Am. A 5, 416–425 (1988).
- H. Y. Chang, C. W. Chen, C. K. Lee, and C. P. Hu, “The tapestry cellular automata phase unwrapping algorithm for nondestructive testing,” J. Opt. Lasers Eng. 30, 487–502 (1998).
- C. W. Chen, H. Y. Chang, and C. K. Lee, “An innovative phase shifting system for nondestructive testing,” J. Chin. Soc. Appl. Mech. 14, 31–39 (1998).
- Y. C. Chen, S. S. Lee, C. M. Lee, C. K. Lee, and G. B. Yeh, “New methodology for measuring highly aberrated wave fronts induced by diffractive optical elements,” in Testing, Packaging, Reliability, and Applications of Semiconductor Lasers IV, M. Fallahi, J. Linden, and S. Wang, eds., Proc. SPIE 3626, 248–259 (1999).
- S. Takahashi, “Multilayer piezoelectric ceramics actuators and their applications,” Jpn. J. Appl Phys. 24, 41–45 (1985).
- Tokin America Inc., “Multilayer piezoelectric actuator,” Insp. Rec. 912–46S-02039, displacement: 14.4 μm at 100 V dc for 17.981-mm length (1991).

## Cited By |
Alert me when this paper is cited |

OSA is able to provide readers links to articles that cite this paper by participating in CrossRef's Cited-By Linking service. CrossRef includes content from more than 3000 publishers and societies. In addition to listing OSA journal articles that cite this paper, citing articles from other participating publishers will also be listed.

« Previous Article | Next Article »

OSA is a member of CrossRef.