Removing Aperture-Induced Artifacts from Fourier Transform Infrared Intensity Values
Applied Optics, Vol. 41, Issue 15, pp. 2831-2839 (2002)
http://dx.doi.org/10.1364/AO.41.002831
Acrobat PDF (151 KB)
Abstract
Two Fourier transform infrared intensity artifacts have been observed at moderately high (0.1 cm^{−1}) spectral resolution: Light reflected off the aperture was double modulated by the interferometer, producing a 2f alias, and the warm (≈310 K) annulus of the aperture seen by a cooled detector resulted in distorted line shapes and anomalous intensities in the fingerprint region. Although the second artifact has been alluded to previously, we report corrections to remove both of these anomalies and to demonstrate the efficacy of these corrections. Prior to correction, integrated-band intensities were found to be in error by up to 12%.
© 2002 Optical Society of America
OCIS Codes
(070.2590) Fourier optics and signal processing : ABCD transforms
(110.1220) Imaging systems : Apertures
(120.3180) Instrumentation, measurement, and metrology : Interferometry
(120.4570) Instrumentation, measurement, and metrology : Optical design of instruments
(300.6320) Spectroscopy : Spectroscopy, high-resolution
(300.6340) Spectroscopy : Spectroscopy, infrared
Citation
Timothy J. Johnson, Robert L. Sams, Thomas A. Blake, Steven W. Sharpe, and Pamela M. Chu, "Removing Aperture-Induced Artifacts from Fourier Transform Infrared Intensity Values," Appl. Opt. 41, 2831-2839 (2002)
http://www.opticsinfobase.org/ao/abstract.cfm?URI=ao-41-15-2831
Sort: Year | Journal | Reset
References
- R. J. Bell, Introductory Fourier Transform Spectroscopy (Academic, New York, 1972).
- P. R. Griffiths and J. A. de Haseth, Fourier Transform Infrared Spectroscopy (Wiley, New York, 1986).
- P. M. Chu, F. R. Guenther, G. C. Rhoderick, and W. J. Lafferty, “The NIST Quantitative Infrared Database,” J. Res. Natl. Inst. Stand. Technol. 104, 59–81 (1999).
- S. W. Sharpe, R. L. Sams, T. J. Johnson, P. M. Chu, G. C. Rhoderick, and F. R. Guenther, “Creation of 0.10 cm^{−1} resolution, quantitative, infrared spectral libraries for gas samples,” in Vibrational Spectroscopy-Based Sensor Systems, S. D. Christensen and A. J. Sedlacek, III, eds. Proc. SPIE 4577, 12–24(2001).
- These data are available at http://www.gases.nist.gov/database.html.
- These data are available at http://nwir.pnl.gov.
- J. W. Johns, “High resolution and the accurate measurement of intensities,” Mikrochim. Acta 111, 171–188 (1987).
- L. Mertz, “Auxiliary computation for Fourier spectrometry,” Inf. Phys. 7, 17–23 (1967).
- D. B. Chase, “Phase correction in FT-IR,” Appl. Spectrosc. 36, 240–244 (1982).
- R. C. M. Learner, A. P. Thorne, and J. W. Brault, “Ghosts and artifacts in Fourier-transform spectrometry,” Appl. Opt. 35, 2947–2954 (1996).
- D. B. Chase, “Nonlinear detector response and FT-IR,” Appl. Spectrosc. 38, 491–494 (1984).
- M. Birk, D. Hausamann, G. Wagner and J. W. Johns, “Determination of line strengths by Fourier-transform spectroscopy,” Appl. Opt. 35, 2971–2985 (1996).
- J. Ballard, R. J. Knight, D. A. Newnham, J. Vander Auwera, M. Herman, G. Di Lonardo, G. Masciarelli, F. M. Nicolaisen, J. A. Beukes, L. K. Christensen, R. McPheat, G. Duxbury, R. Freckleton, and K. P. Shine, “An intercomparison of laboratory measurements of absorption cross-sections and integrated absorption intensities for HCFC-22,” J. Quant. Spectrosc. Radiat. Transfer 66, 109–128 (2000).
- M. C. Abrams, G. C. Toon, and R. A. Schindler, “Practical example of the correction of Fourier-transform spectra for detector non-linearity,” Appl. Opt. 33, 6307–6314 (1994).
- A. Keens and A. Simon, “Corrections of non-linearities in detectors in Fourier transform spectroscopy,” U.S. patent 4,927,269 (22 May 1990).
- R. L. Richardson, Jr., and P. R. Griffiths, “Evaluation of a system for generating quantitatively accurate vapor-phase infrared spectra,” Appl. Spectrosc. 52, 143–153 (1998).
- G. Guelachvili, “Distortions in Fourier spectra and diagnosis,” in Spectrometric Techniques, G. A. Vanasse, ed. (Academic, New York, 1981).
- J. R. Birch, and F. J. J. Clarke, “Fifty categories of ordinate error in Fourier transform spectroscopy,” Spectrosc. Europe 7, 16–22 (1995).
- J. W. Johns, “Thermal artifacts in mid- to far-IR FT spectroscopy,” in Fourier Transform Spectroscopy: New Methods and Applications, Vol. 4 of 1995 OSA Technical Digest Series (Optical Society of America, Washington D.C., 1995), pp. 26–27.
- Certain commercial equipment, instruments, or materials are identified in this paper to foster understanding. Such identification does not imply recommendation or endorsement by the National Institute of Standards and Technology, nor does it imply that the materials or equipment identified are the best available for the purpose.
- T. A. Blake, R. L. Sams, S. W. Sharpe, and S. S. Xantheas are preparing a manuscript called “High resolution infrared spectroscopy of perfluorocyclobutane in the 1200 to 1300 cm^{−1} region.”
- J. E. Bertie, “Specification of components, methods and parameters in Fourier transform spectroscopy by Michelson and related interferometers,” Pure Appl. Chem. 70, 2039–2045 (1998).
- L. P. Giver, C. Chackerian, jr., and M. N. Spencer, “Thermal contamination of BOMEM FTIR line shapes,” in High Resolution Fourier Transform Spectroscopy, Vol. 21 of 1992 OSA Technical Digest Series(Optical Society of America, Washington, D.C., 1992), pp. 96–97.
- H. W. H. M. Jongbloetts, M. J. H. van de Steeg, E. J. C. M. van der Werf, J. H. M. Stoelinga, and P. Wyder, “Spectrum distortion of far-infrared Fourier spectroscopy by multiple reflections between sample and Michelson interferometer,” Inf. Phys. 20, 185–192 (1980).
- J. Ballard, J. J. Remedios, and H. K. Roscoe, “Sample emission effect on spectral parameters,” J. Quant. Spectrosc. Radiat. Transfer 48, 733–741 (1992).
- J. R. Birch and E. A. Nicol, “The removal of detector port radiation effects in power transmission or reflection Fourier transform spectroscopy,” Infrared Phys. 27, 159–165 (1987).
Cited By |
Alert me when this paper is cited |
OSA is able to provide readers links to articles that cite this paper by participating in CrossRef's Cited-By Linking service. CrossRef includes content from more than 3000 publishers and societies. In addition to listing OSA journal articles that cite this paper, citing articles from other participating publishers will also be listed.
« Previous Article | Next Article »
OSA is a member of CrossRef.