The Lloyd moiré interferometer was introduced in Appl. Opt. 6, 1707 ( 1967), as a means of testing interferometrically a path across very large surfaces in oblique to grazing incidence. The width of the path is determined by the optics in use. For any sensible angle of incidence, the Lloyd interference fringes will be very narrowly spaced, but they are transformed into conveniently visible fringes by a moiré grating. The precision of the Lloyd moiré interferometer is limited by the very large angle of incidence and by the relatively poor definition of the moiré fringes. The objective of the present paper is to show an improved Lloyd interferometer technique, which with equally simple instrumentation permits one to vary and increase the sensitivity in testing large surfaces in oblique incidence by decreasing the angle of incidence.
© 1970 Optical Society of America
Original Manuscript: November 24, 1969
Published: August 1, 1970
P. Langenbeck, "Higher-Order Lloyd Interferometer," Appl. Opt. 9, 1838-1841 (1970)
- P. Langenbeck, Appl. Opt. 6, 1707 (1967). [CrossRef] [PubMed]
- A. Lohmann, O. Bryndahl, Appl. Opt. 6, 1934 (1967). [CrossRef] [PubMed]
- G. Scherf, Jena Rev. 2, 107 (1966).
|Alert me when this paper is cited|
OSA is able to provide readers links to articles that cite this paper by participating in CrossRef's Cited-By Linking service. CrossRef includes content from more than 3000 publishers and societies. In addition to listing OSA journal articles that cite this paper, citing articles from other participating publishers will also be listed.