In a normal Michelson interferometer, modulated radiation is transmitted and also reflected back toward the source of the radiation. The back-reflected radiation is generally ignored but it can be exploited to provide a very simple means of modifying an FT-IR spectrometer to measure emission spectra, without the necessity of access to the source or source optics. Radiation from the emission source is directed into the interferometer from the direction of the normal sample location. Modulated radiation is reflected back from the interferometer through the sample area to be measured by the normal detector. It has been noted previously that radiation originating from the sample area can contribute to the observed spectrum. However, this has been regarded as a problem rather than as a means of obtaining useful measurements.
M. A. Ford and R. A. Spragg, "A Simple Method of Measuring Emission Spectra with an FT-IR Spectrometer," Appl. Spectrosc. 40, 715-716 (1986)
References are not available for this paper.
OSA is able to provide readers links to articles that cite this paper by participating in CrossRef's Cited-By Linking service. CrossRef includes content from more than 3000 publishers and societies. In addition to listing OSA journal articles that cite this paper, citing articles from other participating publishers will also be listed.