A prototype, no-moving-parts, plane array infrared spectrograph (PA-IR) capable of routine spectral acquisition in the 3400 to 2000 cm-1 region has been constructed. The instrument includes a continuous source, Czerney-Turner type monochromator system and an infrared camera that incorporates a 320 × 256 pixel InSb focal plane array detector cooled with liquid nitrogen. PA-IR spectra (~3400 to 2550 cm-1) of polystyrene (PS) and poly(ethylene naphthalate) (PEN) films have been obtained at a resolution of ~8 cm-1 with excellent signal-to-noise ratios (SNR). Peak-to-peak noise levels of ~1 × 10-3 absorbance units are observed for single acquisition spectra with 1.5 ms integration times and 17 ms total acquisition times. Integration times as low as 10 μs are possible (with good SNR); however, data acquisition is limited by the frame rate (60 frames/s) of the software acquisition package currently used. In this work, an apertured image of the source is displayed over ~20 rows of the array and the expected square root improvement in SNR is observed when multiple rows and/or frames are averaged. We have also shown that a square root improvement in SNR continues to occur with further signal averaging, providing noise levels as low as 1.5 × 10-5 absorbance units. Several additional advantages and options associated with the PA-IR method are discussed, including time-resolved spectroscopy, real-time monitoring, and spectroscopic mapping.
Douglas L. Elmore, Mei-Wei Tsao, Simon Frisk, D. Bruce Chase, and John F. Rabolt, "Design and Performance of a Planar Array Infrared Spectrograph that Operates in the 3400 to 2000 cm-1 Region," Appl. Spectrosc. 56, 145-149 (2002)
References are not available for this paper.
OSA is able to provide readers links to articles that cite this paper by participating in CrossRef's Cited-By Linking service. CrossRef includes content from more than 3000 publishers and societies. In addition to listing OSA journal articles that cite this paper, citing articles from other participating publishers will also be listed.