## Investigation of single-factor calibration of the wave-number scale in Fourier-transform spectroscopy

JOSA B, Vol. 21, Issue 8, pp. 1543-1550 (2004)

http://dx.doi.org/10.1364/JOSAB.21.001543

Acrobat PDF (341 KB)

### Abstract

We have used the fundamental and frequency-doubled output of a single-frequency tunable laser locked to a precisely known transition in molecular iodine to provide calibration of a Fourier-transform spectrometer (FTS) in the visible and near-ultraviolet regions and to investigate the limiting uncertainty involved in calibrating spectra by using a single multiplicative correction to the entire optical frequency scale. An integrating sphere was used to introduce the laser light as a pseudoincoherent source and provide uniform illumination of the FTS field of view. The sphere also served to combine the laser beams with light from a series of mercury electrodeless discharge lamps containing argon carrier gas at selected pressures. Four strong lines in the spectrum of ^{198}Hg were measured with these lamps to obtain precise wavelengths and argon-pressure-shift coefficients. These lines, emitted from lamps with argon pressures in the range 33 Pa (0.25 Torr) to 1330 Pa (10 Torr), are suitable for future calibration of FT spectra without need for the laser source. The limiting relative uncertainty component in the reported wavelengths is 6.19×10^{−9}, as estimated from observed deviation of the frequency ratios of the calibration lasers from the exact value of 2. The adequacy of a single multiplicative correction factor for the absolute calibration of an individual FT spectrum is supported by our data, at the level of better than a part in 10^{8}.

© 2004 Optical Society of America

**OCIS Codes**

(020.3690) Atomic and molecular physics : Line shapes and shifts

(120.3180) Instrumentation, measurement, and metrology : Interferometry

(120.6200) Instrumentation, measurement, and metrology : Spectrometers and spectroscopic instrumentation

(300.6210) Spectroscopy : Spectroscopy, atomic

(300.6300) Spectroscopy : Spectroscopy, Fourier transforms

**Citation**

Marc L. Salit, Craig J. Sansonetti, Damir Veza, and John C. Travis, "Investigation of single-factor calibration of the wave-number scale in Fourier-transform spectroscopy," J. Opt. Soc. Am. B **21**, 1543-1550 (2004)

http://www.opticsinfobase.org/josab/abstract.cfm?URI=josab-21-8-1543

Sort: Year | Journal | Reset

### References

- G. Horlick and W. K. Yuen, “Atomic spectrochemical measurements with a Fourier transform spectrometer,” Anal. Chem. 47, 775A–781A (1975).
- P. Luc and S. Gerstenkorn, “Fourier transform spectroscopy in the visible and ultraviolet range,” Appl. Opt. 17, 1327–1331 (1978).
- J. W. Brault, “Fourier transform spectrometry in relation to other passive spectrometers,” Philos. Trans. R. Soc. London, Ser. A 307, 503–511 (1982).
- L. M. Faires, “Fourier transforms for analytical atomic spectroscopy,” Anal. Chem. 58, 1023A–1034A (1986).
- A. Thorne, “High-resolution Fourier transform atomic spectrometry,” J. Anal. At. Spectrom. 2, 227–232 (1987).
- A. P. Thorne, “Fourier transform spectrometry in the ultraviolet,” Anal. Chem. 63, 57A–65A (1991).
- R. C. M. Learner and A. P. Thorne, “Wavelength calibration of Fourier-transform emission spectra with applications to Fe I,” J. Opt. Soc. Am. B 5, 2045–2059 (1988).
- A. P. Thorne and R. C. M. Learner, “High resolution FTS of atoms and molecules in the ultra-violet,” Mikrochim. Acta II, 445–448 (1988).
- M. L. Salit, J. C. Travis, and M. R. Winchester, “Practical wavelength calibration considerations for UV-visible Fourier-transform spectroscopy,” Appl. Opt. 35, 2960–2970 (1996).
- V. Kaufman, “Wavelengths, energy levels, and pressure shifts in mercury 198,” J. Opt. Soc. Am. 52, 866–870 (1962).
- C. J. Sansonetti, “Precise measurements of hyperfine components in the spectrum of molecular iodine,” J. Opt. Soc. Am. B 14, 1913–1920 (1997).
- F. C. Fehsenfeld, K. M. Evenson, and H. P. Broida, “Microwave discharge cavities operating at 2450 MHz,” Rev. Sci. Instrum. 36, 294–298 (1963).
- To describe experimental procedures adequately, it is occasionally necessary to identify commercial products by manufacturer. In no instance does such identification imply endorsement by the National Institute of Standards and Technology, nor does it imply that the particular products or equipment are necessarily the best available for the purpose.
- A. P. Thorne, C. J. Harris, I. Wynne-Jones, R. C. M. Learner, and G. Cox, “A Fourier transform spectrometer for the ultraviolet: design and performance,” J. Phys. E 20, 54–60 (1987).
- P. Saarinen and J. Kauppinen, “Spectral line-shape distortions in Michelson interferometers due to off-focus radiation source,” Appl. Opt. 31, 2353–2359 (1992).
- J. Kauppinen and P. Saarinen, “Line-shape distortions in misaligned cube corner interferometers,” Appl. Opt. 31, 69–74 (1992).
- J. W. Brault and M. C. Abrams, “DECOMP: a Fourier transform spectra decomposition program,” High Resolution Fourier Transform Spectroscopy, Vol. 6 of 1989 Technical Digest Series (Optical Society of America, Washington, D.C., 1989), pp. 118–121.
- R. C. M. Learner, A. P. Thorne, I. Wynne-Jones, J. W. Brault, and M. C. Abrams, “Phase correction of emission line Fourier transform spectra,” J. Opt. Soc. Am. A 12, 2165–2171 (1995).
- J. Genest and P. Tremblay, “Instrument line shape of Fourier transform spectrometers: analytic solutions for nonuniformly illuminated off-axis detectors,” Appl. Opt. 38, 5438–5446 (1999).

## Cited By |
Alert me when this paper is cited |

OSA is able to provide readers links to articles that cite this paper by participating in CrossRef's Cited-By Linking service. CrossRef includes content from more than 3000 publishers and societies. In addition to listing OSA journal articles that cite this paper, citing articles from other participating publishers will also be listed.

« Previous Article | Next Article »

OSA is a member of CrossRef.