A novel implementation of broadband cavity enhanced absorption spectroscopy (BBCEAS) has been used to perform sensitive visible wavelength measurements on liquid-phase solutions in a 2 mm cuvette placed at normal incidence to the cavity mirrors. The overall experimental methodology was simple, low cost, and similar to conventional ultraviolet-visible absorption spectroscopy. The cavity was formed by two concave high reflectivity mirrors. Three mirror sets with nominal reflectivities (R) of R = 0.99, 0.9945, and 0.999 were used. The light source consisted of a high intensity red, green, blue, or white LED. The detector was a compact charge-coupled device (CCD) spectrograph. Measurements were made on the representative analytes, Ho3+, and the dyes brilliant blue-R, sudan black, and coumarin 334 in appropriate solvents. Cavity enhancement factors (CEF) of up to 104 passes for the high reflectivity mirrors were obtained. The number of passes was limited by relatively high scattering and absorption losses in the cavity, of ∼1 × 10−2 per pass. Measurements over a wide wavelength range (420–670 nm) were also obtained in a single experiment with the white LED and the R = 0.99 mirror set for Ho3+ and sudan black. The sensitivity of the experimental setup could be determined by calculating the minimum detectable change in the absorption coefficient αmin. The values ranged from 5.1 × 10−5 to 1.2 × 10−3 cm−1. The limit of detection (LOD) for the strong absorber brilliant blue-R was 620 pM. A linear dynamic range of measurements of concentration over about two orders of magnitude was demonstrated. The overall sensitivity of the experimental setup compared very favorably with previous generally more experimentally complex and expensive liquidphase cavity studies. Possible improvements to the technique and its applicability as an analytical tool are discussed.
Vol. 2, Iss. 7 Virtual Journal for Biomedical Optics
Meez Islam, L. Nitin Seetohul, and Zulfiqur Ali, "Liquid-Phase Broadband Cavity-Enhanced Absorption Spectroscopy Measurements in a 2 mm Cuvette," Appl. Spectrosc. 61, 649-658 (2007)
References are not available for this paper.
OSA is able to provide readers links to articles that cite this paper by participating in CrossRef's Cited-By Linking service. CrossRef includes content from more than 3000 publishers and societies. In addition to listing OSA journal articles that cite this paper, citing articles from other participating publishers will also be listed.